575 Series Passive Samplers are now
VOC Chek 575!
New VOC Chek 575 Downloads
SKC VOC Chek 575 Passive Samplers
Miniature VOC Chek 575 samplers collect organic vapors
without the use of a pump. Individual chemicals simply diffuse from
the atmosphere into the sampler at a fixed rate. SKC has scientifically
validated the sampling rate of the VOC Chek 575 Series passive samplers along
with other critical sampling parameters to meet OSHA/ASTM/ANSI requirements.
SKC VOC Chek 575 Series Passive Samplers are identified as a viable alternative
to active sampling in seven OSHA methods.
- Newly designed housing for easier analysis
- No assembly required
- Accurate and reliable — Proven performers
- Save time and money
Collect most compounds with one sampler for up to a full 8-hour workshift
eliminating multiple trips to the worksite
- One sampler + one analysis = money saved
- Economical solvent desorption and GC analysis
- Variety of samplers available
- Only SKC offers a choice of sorbents to better target specific contaminants
- Proper sorbent selection means no reverse diffusion
- Superior collection efficiency
- SKC sorbents have a larger surface area for higher capacities
- Samplers contain >300 mg of sorbent much
more than other brands
- Sampling flexibility
- Validated for 15 minute and 8-hour sampling
- Suitable for 24-hour indoor air sampling for some compounds
- Accurate and reliable
- So easy to use!
- No pump or training required
- Meet ASTM D6246-08 and ANSI/ISEA 104-1998(r 2009) standards
- Identified as viable alternative to active sampling
in seven OSHA methods
- Easy access to sampling rates and parameters in the SKC Passive Sampling Guide
Full-shift Sampling with only ONE
SKC VOC Chek Passive Samplers are validated for a full 8 hours, that's TWICE the
limit value! This assures the integrity of the sampler, so you only
need ONE sampler to get a true 8-hour TWA!
Using a VOC Chek Passive Sampler is easy as 1, 2, 3!
Remove front cap.
Clip in worker's breathing zone.
Cap and record information.
VOC Chek 575 Series Passive Samplers
NIOSH Protocol 1
Passive sampling is quite different from active sampling with sorbent
tubes. With this fact in mind, NIOSH established validation protocols
for evaluating the performance of passive samplers. 1 Protocol
validation was developed so that manufacturers could produce passive
samplers that reliably assess airborne levels.
For compliance sampling, it is recommended that only passive samplers with full or bi-level validation be used. Passive samplers with lower levels of validation may be used only if verified by validated sorbent sample tube methods.
- Passive monitor has passed all NIOSH protocol validation requirements
for sampling rate, desorption efficiency, humidity effects, reverse
diffusion, storage stability test for reactive compounds whose
stability is suspect, and a factorial study that includes interfering
compounds. This is the most rigorous test possible and includes
all parameters that can affect sampling accuracy (formerly referred
to as Level 5).
- A key member of an homologous series has passed full validation
and all other members have passed partial. The validity of this
shown by Guild et al. 2 (formerly
referred to as Level 5B).
- Passive monitor has passed NIOSH protocol requirements for sampling
rate, desorption efficiency, humidity effects, reverse diffusion,
and storage stability test for reactive compounds. Partial validations
are not recommended for compliance sampling.
- Uptake rate that is calculated from the physical properties of
the analyte and geometry of the sampler. Calculated rates are not
based upon direct laboratory testing of the analyte with the sampler.
Calculated validations are not recommended for compliance sampling.
About OSHA Testing
After extensive testing on the sampling rate variations among diffusive samplers, OSHA has released the following methods that specify diffusive (passive) samplers including SKC Cat. No. 575-002:
- Toluene — Method 111 (1998)
- Tetrachloroethylene and Trichloroethylene — Method 1001 (1999)
- Xylenes and Ethylbenzene — Method 1002 (1999)
- 2-Butanone (MEK) and Hexone (MIBK) — Method 1004 (2000)
- Benzene — Method 1005 (2002)
- Butyl acetates — Method 1009 (2007)
- Styrene — Method 1014 (2009)
OSHA continues its work on passive method development based on the premise that diffusive samplers are a viable alternative to tube/pump sampling.
VOC Chek 575 Passive Sampler Ordering Information
instructions are included with each sampler package
| Passive Sampler for Organic Vapor
| Passive Sampler for Organic Vapor
| Passive Sampler for Styrene
Anasorb 747 coated with
SKC Passive Sampler for Ethylene Oxide
- Uses same sorbent as active OSHA Method 1010
- Fully validated and documented performance
- 8-hour and 15-minute sampling
- Small, lightweight, and easy to use — no pump required
- Ideal for hospitals, veterinary clinics, and sterile instrument manufacturers
| Passive Sampler for Ethylene Oxide
|| Anasorb 747 coated with hydrobromic
VOC Chek 575 Passive Sampling Guide for sampling rates
VOC Chek 575 Passive Sampler Product Data Bulletin (PDF)
VOC Chek 575 Passive Samplers Operating/Analysis Instructions (PDF)
Passive Sampler Research Reports
Long-term Indoor/Ambient Air Validation Report
Passive Samplers Meet CEN 838 Diffusive Sampler
List of Laboratories for Analyzing
VOC Chek 575 Passive Samplers
575 Series Passive Samplers Specified in OSHA Diffusive Methods
OSHA specifies SKC VOC Chek 575 Series diffusive (passive) samplers as a
viable alternative to active sampling in six OSHA diffusive methods.
OSHA continues its work on diffusive method development.
Method 111 for Toluene
Method 1001 for Tetrachloroethylene and Trichlorethylene
Method 1002 for Xylenes and Ethylbenzene
Method 1004 for 2-Butanone (MEK) and Hexone (MIBK)
Method 1005 for Benzene
Method 1009 for Butyl Acetates
Method 1014 for Styrene
For a listing of laboratories in the U.S. and Canada that analyze Passive Samplers,
see the 575 Series Passive
Laboratories Reference List.
Custom Passive Samplers Available
For specific applications, SKC will manufacture
samplers to meet your specific requirements. Download the Custom Passive Sampler
Order Form in PDF format. Complete
form and fax it to your SKC representative, contact
your SKC representative, or call the
SKC Sales Department at 724-941-9701 for more information.
- Cassinelli, M.E., Hull, R.D., Crable, J.V. and Teass, A.W., "Diffusive
Sampling: An Alternative to Workplace Air Monitoring," A. Berlin, R.H. Brown
and K.J. Saunders (Royal Society of Chemistry, London) (eds.), NIOSH Protocol
for the Evaluation of Passive Monitors, 1987, pp 190-202.
- Guild, L.V., Myrmel, K.H., Myers, G. and Dietrich, D.F., "Bi-Level Passive
Monitor Validation: A Reliable Way of Assuring Sampling Accuracy for a Larger
Number of Related Chemical Hazards" Appl. Occup. Environ. Hyg., Vol
7, No. 5, May 1992, pp. 310-317. Reprints available
from SKC! Email your mailing address to SKC Inc at email@example.com.
This page was last modified August 26, 2014