GS-3 Respirable Dust Cyclone
Meets ISO 7708/CEN Criteria
GS-3 Cyclone’s Unique Design
- Conductive plastic construction eliminates static collection problems with charged particles.
- Multiple inlets eliminate sensitivity to wind velocity and user orientation to the contaminant source.
The Multiple-inlet GS-3 Cyclone
The GS-3 Cyclone is a 10-mm lightweight conductive plastic sampler used with a standard three-piece cassette with filter for the collection of respirable dust particles. The GS-3 Cyclone’s removable cassette adapter securely holds the filter cassette in place during sampling. Designed to conform to ISO 7708:1995/CEN particle size-selection criteria, the GS-3 Cyclone has a 50% cut-point of 4 µm (bias within ISO/OSHA/NIOSH requirements) at 2.75 L/min (Calibrated at U.K. Health and Safety Laboratory).
OSHA Publishes Final Rule on Respirable Crystalline Silica
In its final silica rule, OSHA sets a new permissible exposure limit (PEL) of 50 µg/m3 across all industries covered by the rule and abandoned the previous silica PEL formula. In addition, the OSHA final silica rule states that any sampler conforming to the ISO 7708/CEN criteria (50% cut-point of 4 µm) can be used for workplace silica. The ISO 7708 criteria have been adopted by NIOSH, ACGIH, and many other global occupational hygiene organizations. SKC GS-3 Cyclone performance data, relative to the ISO 7708/CEN standard, was published in the Journal of Aerosol Science, 28, 1997.
Traditionally, respirable dust sampling with a cyclone has been performed using a clear styrene cassette. NIOSH now suggests that conductive black polypropylene cassettes are a better option for this application to minimize cassette wall losses.
Collection efficiency relative to ISO 7708 standard in OSHA final silica rule and ACGIH TLVs
(click to enlarge)
GS-3 Cyclone Superior Performance
With low mean bias and higher flow rate requirements, the GS-3 Cyclone provides better sampling accuracy when compared to the performance of other cyclones used for respirable dust collection. Furthermore, the multiple-inlet GS-3 Cyclone overcomes sampling problems that have been reported with the single-inlet 10-mm nylon cyclone.
Kar, K. and Gautam, M., "Orientation Bias of the Isolated 10 mm Nylon Cyclone at Low Stream Velocity," AIHA Journal, Vol. 56 (1995), pp. 1090-1098
Gautam, M. and Sreenath, A., "Performance of a Respirable Multi-inlet Cyclone," Journal of Aerosol Science (U.K.), Vol. 28, No. 7 (1997), pp. 1265-1281
Trakumas, S., et al., Performance Assessment of Personal Respirable Cyclone Samplers, AIHce Presentation 191, 2003, https://bit.ly/3GqxJ1C (PowerPoint presentation)
OSHA Final Rule on Respirable Crystalline Silica, https://www.osha.gov/silica/
Contact SKC Technical Support for reprints.
SKC Limited Warranty and Return Policy
SKC products are subject to the SKC Limited Warranty and Return Policy, whichprovides SKC’s sole liability and the buyer’s exclusive remedy. Toview the complete SKC Limited Warranty and Return Policy, go to the SKCLimited Warranty for Products and Return Policy page.